Getting started INFRA S50
Connect your S50 Sound Level Meter(s) with an INFRA Datalogger using any of the available INFRA Sensor Cables.
Start the Datalogger and wait for the initialization process to be finished.
Under the menu Nodes you’ll find your S50 unit(s) as SND S50-, where SND is the customer string and is the serial number.
Select SND S50- in order to choose Standard (S01, S02, etc.). Then, for the chosen standard, select Interval time, Record time and Upper threshold level (value and on/off). It is also possible to change the customer string.
Note! When doing a test or reference measurement, be sure that the Upper threshold is set to OFF. Otherwise unnecessary transients will be recorded.
5 Activate REG ON, and the measurement will be initiated.
Note! First registration of interval data is delayed to synchronize to the local time. The delay may be up to one interval period. First presentation of data will then be delayed up to two interval periods. Transient trig and readout of Live-values are available directly after activation of REG ON.